---
product_id: 61705011
title: "Flowline LU27-41 EchoSonic II Ultrasonic Level Transmitter with 9.8' Cable and Fob, 1\" NPT"
brand: "flowline"
price: "€ 1108.71"
currency: EUR
in_stock: true
url: https://www.desertcart.nl/products/61705011-flowline-lu27-41-echosonic-ii-ultrasonic-level-transmitter-with-9
store_origin: NL
region: Netherlands
---

# Flowline LU27-41 EchoSonic II Ultrasonic Level Transmitter with 9.8' Cable and Fob, 1" NPT

**Brand:** flowline
**Price:** € 1108.71
**Availability:** ✅ In Stock

## Quick Answers

- **What is this?** Flowline LU27-41 EchoSonic II Ultrasonic Level Transmitter with 9.8' Cable and Fob, 1" NPT by flowline
- **How much does it cost?** € 1108.71 with free shipping
- **Is it available?** Yes, in stock and ready to ship
- **Where can I buy it?** [www.desertcart.nl](https://www.desertcart.nl/products/61705011-flowline-lu27-41-echosonic-ii-ultrasonic-level-transmitter-with-9)

## Best For

- flowline enthusiasts

## Why This Product

- Trusted flowline brand quality
- Free international shipping included
- Worldwide delivery with tracking
- 15-day hassle-free returns

## Description

Offered in a type 6P enclosure with cable interface, the general purpose ultrasonic level transmitter provides continuous level measurement up to 32.8' with a 4-20 milliampere signal output, and is configured via our free Webcal software. This non-contact liquid level sensor is ideally suited for corrosive, ultrapure, sticky or dirty liquids, and is broadly selected for bulk storage, day tank, lift station and process tank level applications. Submersible type 6P enclosure is ideal for sump and below grade installations. Short 4" or 8" dead band maximizes the measurable filling capacity of the tank. Measurement range: 4" to 9.8'. Accuracy: +/- 0.2 percent of range. Resolution: 0.019". Dead band: 4". Beam width: 2". Configuration: WebCal PC. Windows USB 2.0. Memory: Non-volatile. Supply voltage: 24 VDC (loop). Consumption: 0.5W. Loop resistance: 500 ohms at 24 VDC. Signal output: 4-20 milliampere two-wire. Signal invert: 4-20 or 20-4 milliampere. Signal fail-safe: 4, 20, 21, 22 milliampere or hold last. Process temprature: -4 degree to 140 degree F. Temprature compensation: automatic. Ambient temprature: -31 degree to 140 degree F. Pressure: MWP equal to 30 PSI. Enclosure rating: Type 6P, encapsulated, corrosion resistant and submersible. Enclosure material: polycarbonate. Transducer material: PVDF. Cable jacket material: polyurethane. Cable type: 4-conductor, shielded. Cable length: 10'. Process mount: 1" NPT. Mount gasket: Viton. Classification: general purpose. Compliance: CE, RoHS. Approvals: cFMus. Measures 4.1" width by 2.9" height.

## Features

- DSP auto adaptive filters enable plug and play operation optimizing signal output filtering and obstacle recognition     Configuration is fast and easy via WebCal software and USB adapter     Narrow 2" or 3" beam width for applications with limited measurement space     PVDF transducer and 6P polycarbonate enclosure for corrosive liquids     Automatic temperature compensation for accurate measurement

## Images

![Flowline LU27-41 EchoSonic II Ultrasonic Level Transmitter with 9.8' Cable and Fob, 1" NPT - Image 1](https://m.media-amazon.com/images/I/51sAFkMJtCL.jpg)

---

## Why Shop on Desertcart?

- 🛒 **Trusted by 1.3+ Million Shoppers** — Serving international shoppers since 2016
- 🌍 **Shop Globally** — Access 737+ million products across 21 categories
- 💰 **No Hidden Fees** — All customs, duties, and taxes included in the price
- 🔄 **15-Day Free Returns** — Hassle-free returns (30 days for PRO members)
- 🔒 **Secure Payments** — Trusted payment options with buyer protection
- ⭐ **TrustPilot Rated 4.5/5** — Based on 8,000+ happy customer reviews

**Shop now:** [https://www.desertcart.nl/products/61705011-flowline-lu27-41-echosonic-ii-ultrasonic-level-transmitter-with-9](https://www.desertcart.nl/products/61705011-flowline-lu27-41-echosonic-ii-ultrasonic-level-transmitter-with-9)

---

*Product available on Desertcart Netherlands*
*Store origin: NL*
*Last updated: 2026-03-12*